Memory Placement Sheath
The Sheath is made of flexible memory material that allows the catheter to comfortably conform to the shape of the cervix, the cervicouterine junction, and the uterus.
Unlike other systems, the thinner diameter of the outer sheath can be placed into the endocervix to the level of the internal os with no additional dilation required, allowing the catheter to pass easily through the cervix and into the uterine cavity.
Soft Catheter Tip
The soft tip of the catheter allows for smooth insertion into the cervical canal, minimizing tearing of cervical tissue.
“Unlike standard catheters, it has features that make HSG and SIS procedures easier, thus faster to perform. In addition, the catheter is more comfortable for my patients and that means there’s a better chance of procedure completion.”
—Charles E. Miller, MD
For administering contrast media or saline during hysterosalpingography and saline infusion sonohysterogram procedures to detect uterine pathology such as polyps, fibroids, adhesions, or endometrial thickening, and/or to examine the patency of the fallopian tubes.
Suspected infection, suspected pregnancy, profuse bleeding, or sexually transmitted disease.
Do not exceed the recommended balloon inflation volume (1.5 cc for the 5 Fr catheter) or the balloon may burst.
The use of OIL-BASED contrast media such as ethyl esters may interact with the balloon of the catheter, causing possible balloon rupture. The use of aqueous contrast media is recommended.
Some patients may have a hypersensitivity to contrast media.
- Remove catheter from pouch.
- Remove and discard the protective sleeve.
- Attach a syringe with either contrast media or saline (not supplied) to the luer connector of the injection line. Inject contrast media or saline through the catheter to remove air. Advance the insertion sheath so that the distal end of the catheter protrudes slightly from the distal end of the sheath.
- View the cervix and advance the sheath and catheter together into the cervix. (Click on Figure 1 to enlarge)
- While holding the sheath, advance the catheter through the cervical canal and into the uterus. When the catheter is blocked by the uterus, follow the steps detailed in Figures 2-5 (Click to enlarge).
- Open the stopcock and slowly inflate the balloon with either air or saline up to 1.5 cc for the 5 Fr catheter (Click on Figure 6 to enlarge).
- Close the stopcock, allowing the balloon to remain inflated. Gently withdraw catheter so balloon will rest against the internal os.
- Inject contrast media or saline into the uterus and complete the examination.
- Open the stopcock to deflate the balloon. Withdraw the catheter.
Sample RequestEmail: email@example.com
|Instructions For Use|
|Patient Infromation Card|
|Professional Healthcare Information Card|